Medical Gas Analysis- Measuring oxygen in Medical Gas
USP37 Medical Gas requires a specific, selective and positive identification test using a paramagnetic signal
Regular analysis of medical gas is required to ensure that the oxygen concentration is NLT 99.0% volume. Following the appropriate United States Pharmacopeia standards, in this case specifically USP37, requires an analyzer which uses a paramagnetic based technology. The Systech Illinois Paramagnetic Oxgyen Analyzer PM700 is perfectly suited for this medical gas analysis application.
USP -NF Standard:
- Medical Oxygen contains not less than (NLT) 99.0% oxygen by volume.
- Acceptance criteria: NLT 99.0% by volume USP37
- Mode: paramagnetic oxygen measurement (see Medical Gases Assay 415)
- Identification: the paramagnetic signal exhibited by the sample gas in the Assay confirms the presence of oxygen USP37.
- Analysis: determine the percentage by volume of oxygen using a suitable paramagnetic analyzer.
- The certified standards called for are listed under Reagents, Indicators and Solutions.
- Zero gas: nitrogen certified standard
- Span gas: oxygen certified standard
European Pharmacopeia 2.5.27 Oxygen in Gases
European Pharmacopeia 2.5.27 dictates that oxygen in gases be determined using a paramagnetic analyzer.
- Oxygen in gases is determined using a paramagnetic analyzer.
- The principle of the method is based on the high paramagnetic sensitivity of the oxygen molecule.
- Oxygen exerts a strong interaction on magnetic fields, which is measured electronically, amplified and converted to a reading of oxygen concentration.
- As the paramagnetic effect of oxygen is linear the instrument must have a suitable range with a readability of 0.1 percent or better.
- The measurement of oxygen concentration is dependent upon the pressure and temperature so if the analyzer does not automatically compensate for variations it must be calibrated immediately prior to use.
Our PM700 Paramagnetic analyzer may be used for the following applications:
- EP monograph of Argon – IDENTIFICATION A. “Verify that the gas is not oxygen using a paramagnetic analyzer (2.5.27).”
- USP 37 Medical Air (19.5% to 23.5%)
- USP 37 Oxygen NLT 99.0% by volume
- Aviator’s Breathing Oxygen (ABO)
- Breathable Air (BA)
- Liquid Oxygen (LOX) Type II Grade A
- CA OSHA 1910.134 CGA Grade D – for SCBA or Airline Respirators